Model © Sears, Roebuck and Co., Hoffman Estates, IL U.S.A. If this CRAFTSMAN Multimeter fails to give complete satisfaction within one year. Owner’s Manual AutoRanging Digital MultiMeter Model No. CAUTION: Read, understand and follow Safety Rules and Operating Instructions in this. Craftsman Digital Mini Multimeter – Voltage Testers – Amazon. com.
|Published (Last):||6 October 2018|
|PDF File Size:||18.27 Mb|
|ePub File Size:||11.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
Troubleshooting Electrical Testing Item No.
Set the function switch to the FREQ position. Multimeter Craftsman Owner’s Manual 36 pages. Disconnect the test leads from the meter and any item under test. Insert the black test lead banana plug into craftskan negative COM jack. As a result, the reading may show 0 volts when the outlet actually has voltage on it. NEVER operate the meter unless the back cover and the battery and fuse doors are in place and fastened securely. To avoid electric shock, never measure continuity on circuits or wires that have voltage on them.
This MultiMeter is designed to provide years of dependable service, if the following care instructions are performed: To avoid electric shock, disconnect both test probes from any source of voltage before making a temperature measurement. For measurement situations requiring that a range be manually selected, perform the following: Read the current in the display.
Read, understand and follow Safety Rules and Operating Instructions in this manual before using this product. Keep the probe touching the part under test until the 823444 stabilizes about 30 seconds. Insert the black test lead banana plug into the negative COM jack and the red test lead banana plug into the positive V jack.
Call our Customer Service Line Reverse the probe polarity by switching probe position. Open the battery door by loosening the screw using a Phillips head screwdriver. By pressing ‘print’ button you will print only current page. Secure with the two screws.
This Multimeter is designed to provide years of dependable service, if the following care instructions are performed.
Set the function switch to position. Put the battery door back in place. Disconnect the test leads from the meter. The probe tips may not draftsman long enough to contact the live parts inside some V outlets for appliances because the contacts are recessed deep in the outlets.
Crafgsman the test probes to the diode under test. If the battery is good and the meter still does not operate, check to be sure that both ends of the fuse are properly installed.
To avoid electric shock, disconnect power to the unit under test and discharge all capacitors before taking any resistance measurements. The digital reading will indicate the proper decimal point, symbols Hz, kHz and value. Tilt stand craftsmzn battery access is on the rear of unit.
Your browser security needs to be updated
Disconnect the test leads from the meter. On some low AC and Craftxman voltage ranges, with the test leads not connected to a device, the display may show a random, changing reading. Read the voltage in the display. Large voltage surges may occur that can damage the meter. Forward voltage will indicate 0. Touch the test probe tips to the diode or semiconductor junction you wish to test. Remove the batteries and unplug the line cords. To avoid electric shock, do not operate your meter until the battery door is in place and fastened securely.
To avoid electric shock, do not operate your meter until the fuse door is in place and fastened securely. Set the function switch to the CAP position.
CRAFTSMAN Calibration – Micro Precision
Install the new fuse into the holder. To avoid electric shock, disconnect test leads from crafsman source of voltage before making a temperature measurement. The reading will stabilize and give a proper measurement when connected to a 83244. To avoid electric shock, do not test any diode that has voltage on it. Enter text from picture: The rules listed below must be carefully followed for safe operation.
The display will indicate the proper decimal point and value.
Perform any measurement as described in the operating instructions. Be sure that the thermocouple has been removed before changing to any other measurement function. High-voltage circuits, both AC and DC, are very dangerous and should be measured with great care.